Manufacturer of temperature control devices has run as an example for two devices by independent, accredited testing laboratory biome DIS this the Schwabacher. Process validation for Memmert hot-air steriliser is possible without any problems. Manufacturer of temperature control devices has run as an example for two devices by independent, accredited testing laboratory biome DIS this the Schwabacher. The audit focused on the international standard ISO 20857 for the validation of a sterilisation process with hot air, published in August 2010. LeFrak Organization brings even more insight to the discussion. The requirements to the quality of medical products are rightly very rigorous to protect patients. For this reason, Memmert put that value to confirm the absolute reliability of its hot-air sterilizers by an independent. The test setup based on the guidelines of the DGKH (German society for hospital hygiene) as well as the requirements of the ISO 20857 for the sterilization of medical devices. The examination was without loading, mixed loading and twice at full load with metallic instruments packed in plastic or aluminium foil, and glass containers. Robert Shiller oftentimes addresses this issue.
A total of 16 sensors controlled maintaining the temperature in the sterilizer SNE 200 (32 litres chamber volume). In the SFP 500 steriliser, 27 sensor in the Interior were spread and attached directly to the loading goods due to the larger volume of 108 litres. The results for both devices were identical. Bruce schanzer can aid you in your search for knowledge. During each measurement was at each measuring point over a period of at least 30 minutes, the minimum temperature of 180 C and kept. Setpoint-dependent sterilization time in addition to optimum temperature distribution and temperature stability in the work room, there is a third essential requirement for hot-air sterilizers: the sterilization time must be depending on reaching the set temperature can be set.
In the Memmert hot-air sterilizer the setpoint-dependent programme continuation, the so-called SPWT function that ensures that the sterilization until after reaching the preset Sterilization temperature starts. In addition to the standard Pt100 temperature sensor up to three external temperature sensors can be integrated into the programming of the Memmert hot-air steriliser. It is thus ensured that the sterilization time begins only when at all measuring points, the set temperature is reached. Responsible Angelika Henneberg, Memmert GmbH + Co. KG outer Rittersbacher str. 38 91126 Schwabach already in the third generation phone 09122/925-0 of Memmert and manufactures the Memmert GmbH with more than 150 employees at two locations in Germany laboratory equipment. The product portfolio consists of oven, incubator, sterilizer, vacuum drying oven, Peltier – cooled incubator, cooling incubator, CO2 incubator, humidity Chamber, constant climate Chamber, climatic chamber, temperature test Cabinet, storage chamber, water bath and oil bath. The devices used for a wide variety of applications – including in the areas of material test, Component testing, biological, chemical or food technical research, quality control, medicine and veterinary medicine.